The Ventilator adopts microcomputer control with high-difinition TFT color LCD display, integrating the functions of respiratory control and displaying the monitoring parameters. It's versatile in function and excellent in performance.The high-precision Vaporizer is designed with a stable concentration output, automatic compensation for pressure, temperature and flow rate. And it's available to choose one kind of gas from Halothane, Enflurance, Isoflurane and Sevoflurane, ensuring a successful operation. High-definition four-tube flowmeter, designed with stoppers for oxygen and nitrous oxide to control the output oxygen concentration. Reliable pneumatic transmission system. Economic and proper configuration of technologies, optional as required by the customer.Upgrading is available as required by the customer.
Part of Versatile Anesthesia Machine:
Main body: High-strength engineering plastic rack,light,beautiful and corrosion resistant
Scope of application: adult,child
Gas sourceO2: 0.27~0.55MPa N2O: 0.27~0.55MPa
Flow meter: O2: 0.1~1.0L/min 1.1~10L/min; N2O: 0.1~1.0L/min 1.1~10L/min
O2, N2O linkage and N2O Stopper
When using nitrous oxide,oxygen concentration>25%.
When the oxygen pressure flow rate of rapid oxygen supply is 25~75L/min, low oxygen pressure alarm.There will be sound alarm when the oxygen pressure<0.2MPaVaporizer: It has the function of automatic compensation based on pressure,temperature and flow rate. The regulation range of evaporator concentration is 0~5 vol%.
Among Halothane, Enflurane, Isoflurance and Sevoflurance, one can be chosen for application as required by the customer.
Vaporizers that are imported with original packaging are also available.Respiratory circuit
Working mode: all-close, semi-clos, semi-open
Releasing pressure: 0.5~7kPaRespiratory bellows
Bellows for adults, bellows for children
Tidal volume range: 0~1,500ml
Part of Ventilator:
Display mode: High-definition TFT color LCD screen display
Ventilation model: PPV, SIPPV, SIMV, PEEP, MANUAL, SIGH
Tidal volume: 50~1500ml(20~1500ml); Rate: 1~99 bpm
SIMV rate: 2~20 bpm; I:E: 2:1~1:8(4:1~1:8)
Inspiratory trigger pressure: -1.0~2.0Kpa; PEEP: 0~2.0Kpa
Pressure Range: 1.0~6.0Kpa; SIGH: 1.5 times the inspiratory time (70~150 adjustable)
Inspiratory platform: 0~50%.Parameters for ventilation monitoringtidal volume, ventilation volume, IPPV rate, SIMV rate, total respiratory rate, I/E, peak pressure of airway, average pressure, pressure-time waveform, flow rate-time waveform, PEEP, inspiratory trigger pressure, Inspiratory platform(antonomous inspiratory rate, lung compliance, airway resistance, flow-volume oop, pressure-volume loop)Monitoring of oxygen concentration21%~100%
Security Alarm System:Oxygen Concentration alarm
upper limit setting range: 21%~100%; Low limit setting range: 10%~80%Airway pressure alarm: upper limit setting range: 0.3~6.0kPa;
Low limit setting range: 0.2~5.0kPa
Per-minute ventilation volume alarm: Upper limit setting range: 3.0~30L/min;
Low limit setting range: 1.0~10L/min
Sustained high-pressure alarm: It will give alarm when the stress is consistently higher than 2.5kPa.
Suffocation alarm: It will give sound and light alarm if there is no tidal volume input for 15 seconds.
Power alarm: Power AC 220V±10% 50Hz±1Hz (UPS, Storage battery)
End-expiratory CO2 monitoring